To make sure your project has the most accurate measurement results, always calibrate (also known as compensate) your FaroArm and FARO® Laser Line Probe (LLP) before you start your measurement project.
- Click here to check the version number of the driver installed on your computer to be sure you use the correct instructions. If the version number is 22.214.171.124 or earlier, use the instructions in this article. If not, click here to use more recent instructions.
- Download the appropriate USB FaroArm Manual for step-by-step instructions about calibrating your FaroArm, ScanArm, or LLP.
- Open the manual and type Calibration in the Search field to find the instructions.
- Watch the video below for an overview of how to do the steps. It is easier to do when you can see how to do it.
NOTE: When you have finished taking the 9-points on the calibration plate with the ball probe, remember to pull back from the plate and click the red button on the handle before continuing to calibrate the laser beam of the LLP.
In this 4-minute YouTube tutorial, watch FARO trainer Michelle Edwards calibrate the FaroArm hard probe and the LLP. Notice how Michelle pulls back from the plate and clicks the red button after taking the 9 points on the calibration plate with the ball probe, before she moves on to compensate the laser beam of the LLP. Cannot access YouTube? Download Video
Please review this table before you begin to ensure your success.
|Time to Complete||15 minutes|
|Compatibility||Click here for details about which Microsoft Windows version supports different driver versions computer processors.|
FARO® CAM2® Measure 10.x, CAM2 Measure Q, CAM2 Measure X, CAM2 4.x
scanarm, scanner, calibrate, calibration, llp calibration, laser line probe calibration, llp settings, laser compensation, compensate, compensation